Description of Bernoullis Experiment :
Bernoulli's Apparatus consists essentially of a two dimensional rectangular section convergent divergent duct designed to fit between Constant Head Inlet Tank and Variable Head Outlet Tank. An eleven tube static pressure manometer bank is attached to the convergent divergent duct. A dye injection system is provided which allows for a single filament of dye to be introduced into the entrance to the convergent section to enable laminar and turbulent flow regimes to be demonstrated. The differential head across the test section can be varied from zero up to a maximum of 450 mm. The test section is manufactured from acrylic sheet. The convergent divergent duct is symmetrical about the centre line with a flat horizontal upper surface into which the eleven static pressure tappings are drilled. The lower surface is at an angle of 4° 29'. The width of the channel is 6.35 mm. The height of the channel at entry and exit is 19.525 mm and the height at the throat is 6.35 mm. The static tappings are at a pitch of 25 mm distributed about the centre and therefore about the throat.
Experimental Capability of Bernoullis Experiment :
Accessories Required of Bernoullis Experiment :
Dimensions And Weights:
Net: 525 x 70 x 500 mm, 5.0 Kg
Gross: 0.026m³, 7.0 Kg